Elemental characterization of defects and contamination on surfaces is a necessity for failure analysis. Energy Dispersive X-Ray Spectrometry (EDS), combined with Scanning Electron Microscopy (SEM) provides a rapid and efficient tool for characterizing particles and defects which are on the order of 1µm or larger. Modern instruments allow for the detection of light elements (B through F), as well as the higher Z elements, and for simultaneous imaging of multiple elements. The lateral distribution of seven elements is shown in Figure 1 along with secondary electron and back-scattered electron images. From this collection of images it is readily determined that the contamination is a conglomeration of carbon bearing material and oxides of Mg, Al and Si.
Figure 1 shows Secondary Electron Image (SEI) of the surface defect and associated EDS elemental images and back-scattered electron image.
For particles and defects smaller than 1 micron, an ideal tool is Auger Electron Spectroscopy (AES). With this instrument, features as small as 200Å can be characterized elementally. The Auger survey spectrum of a 500Å particle on a silicon wafer reveals the particle contains aluminum.
Contact us today for your Characterization of Surface Contamination needs at +1 800-366-3867 or please complete the form below to have an EAG expert contact you.
To enable certain features and improve your experience with us, this site stores cookies on your computer. Please click Continue to provide your authorization and permanently remove this message.